Cerko Lab System Conductometry devices – allow to carry out precise conductometric titrations.
The devices are intended for scientific research institutes, universities, science and development units, quality control laboratories and sanitary supervision departments.
Varying on the version, the device consists of precise syringe microtitrator with replaceable syringes or membrane titrator controlled by microprocessor and conductometric measuring system. For the technical indications the adequate version of the control program is required.
This device can be controlled by PC computer, CerkoLab program or the software located in the controlling microprocessor. CerkoLab devices can work in different modes. In autonomous mode, when programmed before, the measurements are registered in the internal memory. The acquired data can be sent to the computer afterwards. In the classical mode the device works as standard conductometer. The remote mode allows to control the device completely from the computer, owing to RS232C or USB port. The controlling electronics is equipped with a keyboard consisting of seven buttons and a small LCD display.
Measured values and all the measurement settings are visible on the display. The keyboard allows the full control of the system in the autonomous mode and it plays an adjunctive role in the remote mode.
Owing to the fact that the syringe dosing unit is equipped with three way valve, it is possible to collect the titrant from the supply.
Supplementary informational materials:
|Number of measuring channels:||1.|
|Measuring extent μS/cm:||0.1μS/cm …1S/cm, 1Ω⋅cm … 10MΩ⋅cm.|
|The change of measure extent:||automatic.|
|Temperature measuring extent:||0…100oC +/-0.097oC.|
|Conductivity measuring cell:||any, adequate to the extent with BNC port|
|Temperature measuring probe:||PT1000 sensor.|
|Number of measuring steps:||1…1000.|
|Pause between the measurements:||1sec… 4.5h|
|Dosing unit:||syringe dosing unit or membrane pump.|
|Dosing tubes:||teflon, inner diameter 0.5mm.|
|minimum dose of titrant:||less than 0.01ml (for Hamilton 5ml syringe).|
|syringe capacity:||5ml, 2ml, 1ml, 0.5ml (recommended syringes: Hamilton).|
|number of syringes:||any.|
|flow of titrant:||through three-way valve.|
|minimal dose of titrant:||0.01ml.|
|Cell calibration:||multipoint (min. 1 measuring point).|
|Temperature sensor calibration:||multipoint (min. 2 measuring points).|
|Number of cells:||any.|
|Types of measurements:||titatrion, kinetics, calibration, specialised designations.|
|Inner memory (option):||512kB, 60 series each 1000 points.|
|Measurement file format:||POM, XLS, CSV, HTML, XML.|
|Terminal to computer:||RS232C or USB port.|
|Operating system:||Windows 95/98/Me/NT/2000/XP/Vista/Win7/Win8, Linux(emulator wine required).|
|Software||CerkoLab, Basic or Expert version.|
|Power supply:||external power supply, stabilised +12V/1000mA.|